The working bench use of purification of one hundred grades:
One hundred purification one the intersection of laminar flow and apparatus in working bench, have high-efficient clean close operations area, cupboard of air that filter, it is usually as a part of clean laboratory. In recent years, the already extensive operating environment used for needing clean air in biomedical research, pharmacy, space science,etc. of the extremely net working bench.
Through the filtration of the high-efficient filter, the ultra net working bench can get rid of the pollution suspending the particle, bacterium, mould in the air effectively, make the result of check operation offer better result for the fact that clinical in order to can be believed.
The operation method of one hundred grades of working bench of purification:
1,Put through the power
2,Turn on the air blower in 20 minutes ahead of time.
3,The purple exterior light sterilizes in beginning a theatrical performance in 30 minutes ahead of time.
One hundred grades of working bench of purification use attentive matters:
1,Ones that install newly or for a long time before the unused ultra net working bench is used: Must use the ultra net vacuum cleaner or not have fibrous tool to carry on to 100 grades of working bench and surrounding environment of purification clean.
2,It is forbidden to record on the mesa of job, must try hard to avoid making the movements that obviously upsets the air current while working.
3,At the one hundred the intersection of purification and the intersection of job and mesa, Wuhan wind drench the intersection of room and article need it is unnecessary to be depositedding, in order to keep the clean air current in the workspace flowing normally, without interruption.
Working bench advantage and structure of purification of one hundred grades:
The advantage of purifying the operation platform is easy to operate freely, more comfortable, operation is high in efficiency, prepare time short, start the machine for more than 10 minutes namely operability, can basically use at any time. In factory produce, it inoculates operation to be very large in quantity, need frequently for a long time operate, it is a very ideal apparatus to purify the platform. Purifying the platform is done the pneumatic strength of drum by the three-phase electrical machinery, power about 145- 260W, passes the air by the purpose-built capillary foamed plastics a slice of range upon range of ” super filters ” made up jointly And then blow out, aseptic purification air laminar flow of dust does not to form the successive one, namely as it always goes, ” The high-efficient special air ” ,It removes dust of greater than 0.31m, fungi and bacterium spore,etc.. Purify velocity of flow of air for 24- 30m/min, this prevent pollution that nearby air might harass and cause from enough already, such a velocity of flow will not hinder adopting alcohol burner or firing and sterilizing to the apparatus,etc. of Bunsen burner either. Personnel operate operation under such a aseptic condition, keep the aseptic material not being polluted in the course of shifting and inoculating. Operate, meet, lose electrical power midway, exposed to, filter material in the air, escape by luck pollution while being difficult. Should finish operation rapidly at this moment, and make marks on the bottle, if material inside is at proliferating stage, will be no longer used as proliferating and changing over to and taking root to train afterwards. If is general production materials, because extremely abundant can also remove. For instance can plant and use after waiting until during the course of taking root.
Purify a power of operation and adopt more three-phase Line four systems, among them there is a Line zero, connect the outer cover of machine, should connect firm on the line of ground, in addition the strategic hinterlands of China are all lines of looks, the operation voltage is 380V. The the strategic hinterland of China has certain orders in the circuit to insert, such as the end of a thread has been connected by mistake, the wind chance overturns, the sound is normal or slightly abnormal at this moment, it is calm to purify a front(can observe sound with the alcohol lights flame, should not try for a long time) ,Should cut off power in time, exchange at position answer and then among them any double-phase end of a thread, can solve. Like only inserting two phase, or there is a phase that keep in touch in three phases badly in the three-phase line, then the machine sound is very abnormal, should cut off the power to overhaul carefully immediately, otherwise will burn the electrical machinery.
Stars instrument and equipment of Shangyu Co., Ltd. lies in Zhejiang Province to continue and revitalize a market town in Shangyu district of the city, it is one that collects production, the research and development of products, processing, the enterprises as an organic whole in trade. The principal products of business of our company: The series of the drying cabinet, train case series, the muffle furnace series in the laboratory, distilled water device series, and clean working bench, the wind drenches the room, ventilates the cupboard, complete set of purification equipment such as the bio-safety cupboard.
Our company has stronger technology research and development teams, the strong service system of guarantee after sale, it is a leading person in the complete sets manufacturer of laboratory of China, welcome the masses of customers to choose.