HH-S number shows a pot of product introduction of thermostatical oil bath:
It is an auxiliary thermostatical laboratory apparatus that HH-S counts and shows the thermostatical oil bath pan, and control the temperature by the machine case, thermostatical oil groove the part of the device three make up. The thermostatical trough selects to make of stainless steel, have high temperature resistant, corrosion-resistant characteristics. The machine case is gushed out and moulded precisely with the static of the high-quality cold-rolled stencil plate. Control the temperature device adopt intellectual the intersection of temperature control and instrument, it is the high performance, high-accuracy temperature control module. It features: PID temperature control, it is accurate to control; LED digital display, ocular and accurate; Have advantage such as being high, easy to use of stable and reliable, precision. The instrument of this is suitable for what has been made and distilled, dried, concentrated, warm and steeped chemical medicines or biologicals and heated at constant temperature in every laboratory is used.
Second, HH-S number show one pot of technical parameters of thermostatical oil bath:
1,Power: 220V 50Hz
2,Heating power: 2000W
3,The range of controlling the temperature: Room temperature – 300 degrees centigrades are apparent
4,The size of operating room: 28X28X30(CM)
Third, HH-S number show a pot of operation instructions of thermostatical oil bath:
1,Can refuel according to the capacity of the need while using, in order to shorten the time to heat. (It can’t be too much to pay attention to refueling, in case that the oil is overflowed outside the pan at the time of the boiling; Notice in the pan oil quantity can’t be too low either, can’t make the electric heat in the pan in charge of emerging the oil, the electric heat is in charge of so as not to burn out, cause the oil spill) . Put through the power, open the switch, the indicator lamp is on.
2,Presume the temperature:
(1). Arrange, examine warm normally at press SET, arrange number, in charge of data flicker under the key ‘) Show the instrument enters the temperature to presume the state, presses The presuming value of key increases, press The presuming value is reduced, press and increase the key or will change to reduce the presuming value of key fast continuously, press SET key instrument again and get back to the temperature of normal working state to finish presuming.
(2). The revision of the error: Can revise the revealing value when the value of confirming the instrument reveals is not the correct measurement value, press SET key enter instrument layers of menu 3 second, selected parameter cooperate behind the SC Or The key can revise this parameter, the range of revision of the sensor error gets to